Biologists that use, or are interested in using, microscope cameras and don’t speak engineering.
Clarity on the relevance of camera specs to biological experimentation.
|Hamamatsu listing||Synonyms used by other vendors|
|Full well capacity||Pixel well depth|
|Full well capacity EM-CCD readout||Shift register well depth, Gain register pixel well depth|
|A/D conversion||A/D conversion factor, Digitization, Digital output format|
As with many scientific questions, the answer to the seemingly simple question, “Will I be able to differentiate 1500 molecules from 1475,” is not straightforward. It depends on the photon emission properties of the fluorophore, the number of fluorophores per molecule, and the camera. The heart of this question is understanding the detectable signal change that your camera is able to detect for your specific experiment. And to understand detectable signal change, we need to look at full well capacity.
Detectable signal change (DSC) describes sensitivity to change. For the case of a microscope camera, DSC describes the measureable precision of a change in the signal, making total signal acquired the limiting factor. Full well capacity—which defines the maximum amount of signal the camera can detect—is the most important feature and camera noise is of minimal importance.
Back to the question of detecting 1500 molecules versus 1475, the question becomes, “can I detect a 2 % difference?” Using the following equations for DSC and % DSC:
and solving for S:
Our camera would need to be able to have enough full well capacity to capture 2500 electrons before saturating.
Of course, our sample would need to produce enough photons to create that many electrons in the detector in the time allotted by the exposure and the quantum efficiency of the system.
But what if you wanted to detect this 2 % difference while also keeping other features in the image visible (above noise) and non-saturating. For example, if you were trying to image neurons, where the cell bodies are bright, but the dendrites are much dimmer. Now you need to consider the dynamic range (DR) of the camera.
The total DR of the camera is also a function of the full well capacity, but here camera noise is important to consider. Note that DR does not consider the sample, it is only a property of the camera.
DR is defined as the maximum achievable signal divided by the camera noise.
And is sometimes presented in decibels instead of as a ratio:
The maximum achievable signal is determined by the full well capacity—if the full well capacity is 30,000 electrons, then the maximum number of photons you can collect in one image for that pixel is 30,000. Of course, because not every photon is converted into an electron, this reflects only a fraction of the actual number of photons and is described by the Quantum Efficiency (QE) of the camera at that particular wavelength.
And noise is a combination of read noise, dark noise, and shot noise (discussed in greater detail here). Typically, you can calculate dynamic range using the full well capacity and the read noise. For most biological experiments, dark noise is negligible and only affects the dynamic range at exposures longer than five minutes, and at high signal shot noise is also negligible.
Using the ORCA-Flash4.0 as an example, we divide the full well capacity of 30,000 electrons by the read noise at slow scan, which is 0.9 electrons, to get a dynamic range of 33,000: 1, or 90.4 Db.
Note that for EM-CCDs, when using the EM register dynamic range can be reduced. Once the full well capacity of the EM gain register exceeds the amplified full well capacity of the individual pixel, dynamic range decreases. The net result is a tradeoff between high EM gain and wide dynamic range.
It’s important to recognize that bit depth and dynamic range are not synonymous. Full well capacity divided by bit depth establishes the limit of precision for each gray level—it gives the value for how many electrons per gray level (digitizer bit).
For quantitative image analysis and feature classification, high bit depth is essential.
NEXT SECTION: My samples have very low signal; can this camera accurately image my sample?
It looks like you're in the . If this is not your location, please select the correct region or country below.
You're headed to Hamamatsu Photonics website for JP (Japanese). If you want to view an other country's site, the optimized information will be provided by selecting options below.
For modern websites to work according to visitor’s expectations, they need to collect certain basic information about visitors. To do this, a site will create small text files which are placed on visitor’s devices (computer or mobile) - these files are known as cookies when you access a website. Cookies are used in order to make websites function and work efficiently. Cookies are uniquely assigned to each visitor and can only be read by a web server in the domain that issued the cookie to the visitor. Cookies cannot be used to run programs or deliver viruses to a visitor’s device.
Cookies do various jobs which make the visitor’s experience of the internet much smoother and more interactive. For instance, cookies are used to remember the visitor’s preferences on sites they visit often, to remember language preference and to help navigate between pages more efficiently. Much, though not all, of the data collected is anonymous, though some of it is designed to detect browsing patterns and approximate geographical location to improve the visitor experience.
Certain type of cookies may require the data subject’s consent before storing them on the computer.
This website uses two types of cookies:
There are two ways to manage cookie preferences.
If you wish to restrict or block web browser cookies which are set on your device then you can do this through your browser settings; the Help function within your browser should tell you how. Alternatively, you may wish to visit www.aboutcookies.org, which contains comprehensive information on how to do this on a wide variety of desktop browsers.
Occasionally, we may use internet tags (also known as action tags, single-pixel GIFs, clear GIFs, invisible GIFs and 1-by-1 GIFs) at this site and may deploy these tags/cookies through a third-party advertising partner or a web analytical service partner which may be located and store the respective information (including your IP-address) in a foreign country. These tags/cookies are placed on both online advertisements that bring users to this site and on different pages of this site. We use this technology to measure the visitors' responses to our sites and the effectiveness of our advertising campaigns (including how many times a page is opened and which information is consulted) as well as to evaluate your use of this website. The third-party partner or the web analytical service partner may be able to collect data about visitors to our and other sites because of these internet tags/cookies, may compose reports regarding the website’s activity for us and may provide further services which are related to the use of the website and the internet. They may provide such information to other parties if there is a legal requirement that they do so, or if they hire the other parties to process information on their behalf.
If you would like more information about web tags and cookies associated with on-line advertising or to opt-out of third-party collection of this information, please visit the Network Advertising Initiative website http://www.networkadvertising.org.
We use third-party cookies (such as Google Analytics) to track visitors on our website, to get reports about how visitors use the website and to inform, optimize and serve ads based on someone's past visits to our website.
You may opt-out of Google Analytics cookies by the websites provided by Google:
We inform you that in such case you will not be able to wholly use all functions of our website.