When providing scientific cameras appropriate for the needs of the customer, a traditional trend was to use ultra-sensitive EM-CCD (electron multiplying CCD) cameras for measurements in a very low light region, and sCMOS (scientific CMOS) cameras with large pixel numbers, high speed, and high dynamic range in relatively bright region. In the meantime, CMOS technology continues to develop particularly in terms of noise characteristics, and we have released a camera equipped with the qCMOS (quantitative CMOS) sensor, a new sensor that marks the dawn of a new era in image sensors.
The qCMOS camera is positioned as an ultra-sensitive camera that offers the ultimate in quantitative imaging because of its extremely low noise performance. Therefore, when comparing a qCMOS and EM-CCD camera it is necessary to judge which camera is best suited to your application. The purpose of this article is to compare qCMOS and EM-CCD cameras to help you choose the best camera for your application.
This section describes processes from a signal acquisition to the subsequent digital output in order to compare sensor technologies and the photon detection performances of a qCMOS and EM-CCD camera.
Figure 1 shows the process from the signal acquisition to the digital output in a camera. Photons incident on the sensor are converted to photoelectrons with a probability according to the quantum efficiency of the sensor. The photoelectrons are then converted to voltage and amplified by a floating diffusion amplifier (FDA), and subsequently converted to a digital value by an analog-to-digital converter (ADC) and finally the digital value is output.
In scientific cameras, noise characteristics in this process greatly affect measurements in the very low light region. In particular, it is very important for an ultra-sensitive camera to minimize the readout noise as much as possible in the digital conversion of photoelectron signals.
Figure 1. The process from a signal acquisition to a digital output
Figure 2. Readout process and photon number resolving in a qCMOS camera
Alternatively, with the EM-CCD camera, although the readout noise itself is very large, the photoelectron signal is increased by an electron multiplication mechanism (as shown in Figure 1.) to relatively reduce the readout noise compared to the signal with gain as shown in Figure 3.
However, because the multiplication process in a detector always produces a large fluctuation in the signal called excess noise, the EM-CCD cannot distinguish the original photoelectron number even if the readout noise is relatively minimized compared to the signal. Therefore, although an EM-CCD camera can detect single-photon level signals with low readout noise by using electron multiplication, quantitative properties are lost due to the excess noise.
Figure 3. Readout process and photon number resolving in an EM-CCD camera
Table 1 summarizes the feature comparison of qCMOS and EM-CCD cameras described in this section.
|qCMOS camera||EM-CCD camera|
|Readout noise||<0.3 electron rms||<0.1 electron with EM gain|
|Electron multiplication||Not used||Used|
|Photon detection performance||0, 1, 2, 3, …||0 or more than 1|
Table 1. Feature comparison of a qCMOS and EM-CCD camera
In addition, an example of actual imaging is shown below as a comparison between the qCMOS camera and the EM-CCD camera.
It can be seen that the qCMOS camera, which does not use electron multiplication, has less image fluctuation from frame to frame compared to the EM-CCD camera, resulting in highly quantitative imaging.
This section compares qCMOS and EM-CCD cameras by the SNR (Signal-to-Noise ratio), which is typically used in quantitative discussions of camera performance.
Figure 4 shows the formula for SNR. (Dark current and clock induced charge (CIC) noise in EM-CCDs are omitted here.)
Figure 4. SNR formula
Let’s look at the case of equal light intensity per pixel (Figure 5). This case assumes that the optimum optical system is constructed for the pixel size of each camera. In CMOS cameras, the SNR also decreases as the light intensity decreases because the readout noise is constant regardless of the light intensity. On the other hand, an EM-CCD camera uses multiplication to minimize the readout noise, so the SNR varies little with light level, but the SNR is smaller than that of a CMOS camera in bright region regions due to the constant presence of multiplication fluctuations. Therefore, the traditional trend was that sCMOS cameras are superior in the bright region, while EM-CCD cameras are superior in low light region. However, with the recent development of CMOS technology, qCMOS cameras show SNR comparable to that of EM-CCD cameras in the extremely low light region even at 0.1 photons/pixel/frame.
Figure 5. The SNR comparison when the light intensity is equal per pixel
Next, let’s look at the case of equal light intensity per sensor area (Figure 6). This case assumes that the camera is replaced with the optics unchanged. Since a CMOS camera has a smaller pixel size than the EM-CCD camera and thus the amount of light per pixel is smaller, this comparison includes SNR data of the CMOS cameras with a binning function to input the equivalent amount of light into the pixel unit compared to an EM-CCD.
Figure 6. The SNR comparison when the light intensity is equal per sensor area
Table 2 shows the comparison conditions and the actual cases that the conditions are suited.
Due to the architecture of CMOS sensors, when NXN binning is performed the readout noise per binned pixel increases N times larger compared to that of the single pixel. Therefore, when replacing a camera with a larger pixel size with a qCMOS camera, it is recommended to construct an optical system optimized for the pixel size of the qCMOS camera without binning, to maximize its performance for a fair comparison.
|Condition||Equal light intensity per pixel||Equal light intensity per sensor area|
|Case||When an optical system is optimized for the pixel size of the camera||When the camera is replaced with the optics unchanged.|
Table 2. SNR comparison conditions and the corresponding cases
When comparing qCMOS cameras with EM-CCD cameras, one of the advantages of qCMOS cameras is the versatility. qCMOS cameras are a hybrid camera that combine the advantages of conventional sCMOS cameras with the extremely low noise equivalent to that of EM-CCDs. The following is a list of applications where qCMOS cameras can be used.
Figure 7. Simulation result of test chart sample in low light region
An EM-CCD camera uses electron multiplication to minimize the readout noise as much as possible, so that there is almost no noise in the dark region when a background light is absent, as shown in Figure 7. However, in a realistic case where a background light is present, the background light is also multiplied, resulting in a noticeable noise in the dark region.
For applications in such extremely low-light regions, imaging is generally performed with very long-time exposure to achieve observable SNR such as more than 1 photons/pixel/frame. In this case, EM-CCD cameras still have an advantage over qCMOS cameras for imaging in such extremely low-light regions with very long-time exposure because of the extremely low dark current achieved by cooling performance of the sensor.
EM-CCD is a mature sensor technology that is losing its advantage due to the continuous development of CMOS technology. However, it is still necessary to judge whether a qCMOS camera or an EM-CCD camera has a better SNR depending on the light level and exposure times in experiments.
The qCMOS camera is a hybrid camera that combines the advantages of a conventional sCMOS camera with the extreme low noise equivalent to an EM-CCD camera, therefore the qCMOS camera will bring new breakthrough in conventional scientific measurement.
The qCMOS sensor will get even better with the continuous development of CMOS technology. We hope this article will be a starting point for customers to understand the new technology of the qCMOS camera and to select the best detector by comparing its performance among other photon counting detectors such as EM-CCDs.
It looks like you're in the . If this is not your location, please select the correct region or country below.
You're headed to Hamamatsu Photonics website for US (English). If you want to view an other country's site, the optimized information will be provided by selecting options below.
For modern websites to work according to visitor’s expectations, they need to collect certain basic information about visitors. To do this, a site will create small text files which are placed on visitor’s devices (computer or mobile) - these files are known as cookies when you access a website. Cookies are used in order to make websites function and work efficiently. Cookies are uniquely assigned to each visitor and can only be read by a web server in the domain that issued the cookie to the visitor. Cookies cannot be used to run programs or deliver viruses to a visitor’s device.
Cookies do various jobs which make the visitor’s experience of the internet much smoother and more interactive. For instance, cookies are used to remember the visitor’s preferences on sites they visit often, to remember language preference and to help navigate between pages more efficiently. Much, though not all, of the data collected is anonymous, though some of it is designed to detect browsing patterns and approximate geographical location to improve the visitor experience.
Certain type of cookies may require the data subject’s consent before storing them on the computer.
This website uses two types of cookies:
There are two ways to manage cookie preferences.
If you wish to restrict or block web browser cookies which are set on your device then you can do this through your browser settings; the Help function within your browser should tell you how. Alternatively, you may wish to visit www.aboutcookies.org, which contains comprehensive information on how to do this on a wide variety of desktop browsers.
Occasionally, we may use internet tags (also known as action tags, single-pixel GIFs, clear GIFs, invisible GIFs and 1-by-1 GIFs) at this site and may deploy these tags/cookies through a third-party advertising partner or a web analytical service partner which may be located and store the respective information (including your IP-address) in a foreign country. These tags/cookies are placed on both online advertisements that bring users to this site and on different pages of this site. We use this technology to measure the visitors' responses to our sites and the effectiveness of our advertising campaigns (including how many times a page is opened and which information is consulted) as well as to evaluate your use of this website. The third-party partner or the web analytical service partner may be able to collect data about visitors to our and other sites because of these internet tags/cookies, may compose reports regarding the website’s activity for us and may provide further services which are related to the use of the website and the internet. They may provide such information to other parties if there is a legal requirement that they do so, or if they hire the other parties to process information on their behalf.
If you would like more information about web tags and cookies associated with on-line advertising or to opt-out of third-party collection of this information, please visit the Network Advertising Initiative website http://www.networkadvertising.org.
We use third-party cookies (such as Google Analytics) to track visitors on our website, to get reports about how visitors use the website and to inform, optimize and serve ads based on someone's past visits to our website.
You may opt-out of Google Analytics cookies by the websites provided by Google:
We inform you that in such case you will not be able to wholly use all functions of our website.