Biologists that use, or are interested in using, microscope cameras and don’t speak engineering
Clarity on the relevance of camera specs to biological experimentation
|Hamamatsu listing||Synonyms used by other vendors|
|Effective pixels||Active pixels, Resolution|
|Cell size||Pixel size|
|Effective area||Image area, Active area, Sensor/diagonal|
Reading a camera specifications table can involve not only having to translate from engineering into biology, but from the language of one camera manufacturer to another. This is particularly true of the terms that go into describing how much resolution is achievable.
Let’s start with the terms effective pixels, active pixels, and resolution. These all refer to the number of pixels that go into capturing an image—the square or rectangular grid that you can lay over your image, where each pixel reports on photons captured over a specific area (see figure below).
The area of the image that each pixel covers depends on the size of the pixel and is listed in the specifications table as cell size or pixel size.
Multiplying the cell/pixel size by the number of pixels converts the field-of-view captured in a single image from pixels to millimeters, and is listed as effective area, image area, active area, or sensor format/diagonal.
Why are the number of pixels described as effective pixels or active pixels instead of simply pixels? The words effective and active are used to remove any ambiguity on exactly how many pixels are actually used to capture an image—for some cameras not all the pixels on the sensor contribute to the image, but rather are used for transferring or holding photoelectrons or voltage.
How does the pixel size spec translate into camera resolution?
For starters, it’s important to bear in mind that the optical system limits resolution—if the microscope can’t resolve the image then the camera will not be able to see it either. So we begin by looking at microscope resolution and calculating the size of the projected image of the object to be resolved. And to do this, we need to first define what we mean by resolution.
How is resolution defined?
In 1873, the physicist Ernst Abbe first defined the diffraction limit of an objective as l/2NA. Lord Rayleigh later refined this equation for microscopes to be 0.61 × l/NA—the separation necessary to distinguish two Airy patterns as separate objects.
From Rayleigh’s equation we can calculate the size of the smallest item the objective can resolve, and when this number is multiplied by the magnification of the objective (assuming no additional intermediate optics) we can calculate how large the item will appear to be when it is projected onto the camera’s sensor.
What size pixels do I need to see this level of resolution?
Now that we have determined the resolution of the optical system and the projected size of objects, we can determine the maximum acceptable pixel size needed to meet this resolution. The question becomes one of sampling—given the need to fulfill the Nyquist criteria, which provides a theoretical framework for the minimum number of pixels needed to accurately define an object, we need to ensure that the object of interest is projected on to at least two pixels on the camera sensor (ideally more, 2.5 to 3 is recommended for high resolution microscopy.)
As an example, we’ll imagine trying to resolve peroxisomes from rat liver parenchymal cells that are roughly 0.7 μm in diameter. To resolve a 0.7 μm object, we’d need an objective of similar or higher resolution. From Table 1, the minimum magnification needed would be the objective with10× magnification and an NA of 0.45 (of course, if a wide field-of-view is not needed, most microscopists would oversample and use a higher magnification to achieve a more visually pleasing and informative image). At this magnification, the projected image would be 7.5 μm, so we’d need a camera with a pixel size of 7.5 μm / 2 = 3.7 μm or smaller.
|Objective||Numerical Aperture (NA)||Resolution Limit*(Rayleigh’s criterion) (μm)||Projected Size (μm)||Maximum Acceptable Pixel Size (μm)|
|*Assume 550 nm light|
It looks like you're in the . If this is not your location, please select the correct region or country below.
You're headed to Hamamatsu Photonics website for GB (English). If you want to view an other country's site, the optimized information will be provided by selecting options below.
For modern websites to work according to visitor’s expectations, they need to collect certain basic information about visitors. To do this, a site will create small text files which are placed on visitor’s devices (computer or mobile) - these files are known as cookies when you access a website. Cookies are used in order to make websites function and work efficiently. Cookies are uniquely assigned to each visitor and can only be read by a web server in the domain that issued the cookie to the visitor. Cookies cannot be used to run programs or deliver viruses to a visitor’s device.
Cookies do various jobs which make the visitor’s experience of the internet much smoother and more interactive. For instance, cookies are used to remember the visitor’s preferences on sites they visit often, to remember language preference and to help navigate between pages more efficiently. Much, though not all, of the data collected is anonymous, though some of it is designed to detect browsing patterns and approximate geographical location to improve the visitor experience.
Certain type of cookies may require the data subject’s consent before storing them on the computer.
This website uses two types of cookies:
There are two ways to manage cookie preferences.
If you wish to restrict or block web browser cookies which are set on your device then you can do this through your browser settings; the Help function within your browser should tell you how. Alternatively, you may wish to visit www.aboutcookies.org, which contains comprehensive information on how to do this on a wide variety of desktop browsers.
Occasionally, we may use internet tags (also known as action tags, single-pixel GIFs, clear GIFs, invisible GIFs and 1-by-1 GIFs) at this site and may deploy these tags/cookies through a third-party advertising partner or a web analytical service partner which may be located and store the respective information (including your IP-address) in a foreign country. These tags/cookies are placed on both online advertisements that bring users to this site and on different pages of this site. We use this technology to measure the visitors' responses to our sites and the effectiveness of our advertising campaigns (including how many times a page is opened and which information is consulted) as well as to evaluate your use of this website. The third-party partner or the web analytical service partner may be able to collect data about visitors to our and other sites because of these internet tags/cookies, may compose reports regarding the website’s activity for us and may provide further services which are related to the use of the website and the internet. They may provide such information to other parties if there is a legal requirement that they do so, or if they hire the other parties to process information on their behalf.
If you would like more information about web tags and cookies associated with on-line advertising or to opt-out of third-party collection of this information, please visit the Network Advertising Initiative website http://www.networkadvertising.org.
We use third-party cookies (such as Google Analytics) to track visitors on our website, to get reports about how visitors use the website and to inform, optimize and serve ads based on someone's past visits to our website.
You may opt-out of Google Analytics cookies by the websites provided by Google:
We inform you that in such case you will not be able to wholly use all functions of our website.